The classic laboratory method of measuring high level moisture in solid or semi-solid materials is loss on drying (LOD). In this technique a sample of material is weighed, heated in an oven for an appropriate period, cooled in the dry atmosphere of a desiccator, and then reweighed. If the volatile content of the solid is primarily water, the LOD technique gives a good measure of moisture content. Because the manual laboratory method is relatively slow, automated moisture analyzers have been developed that can reduce the time necessary for a test from a couple hours to just a few minutes. These analyzers incorporate an electronic balance with a sample tray and surrounding heating element. Under microprocessor control the sample can be heated rapidly and a result computed prior to the completion of the process, based on the moisture loss rate, known as a drying curve.
We typically report moisture as a percent weight of total composition. These tests are required for compliance testing as moisture content must be known to properly calibrate potency measurements. This is not to evaluate the quality of your cure; terpene analysis would be more effective.